Recalls and alerts more than 4 years old are automatically archived. While this information can still be accessed in the database, it has not been altered or updated since it was archived. Web pages that are archived on the Web are not subject to the Government of Canada Web Standards. As per the Communications Policy of the Government of Canada, you can request alternate formats by contacting us.
Possible fire hazard with Formula kidney dialysis machines manufactured by Bellco
- Starting date:
- October 20, 2016
- Type of communication:
- Miscellaneous, Medical Device
- Source of recall:
- Health Canada
- Physical Hazard, Medical Devices
- Identification number:
Health Canada is advising Canadians that Formula kidney dialysis machines manufactured by Bellco may pose a potential fire hazard. According to the company, the chance of a fire occurring is low and patients should continue using their machines for dialysis treatment until service technicians can take the appropriate safety precautions.
The following machines are affected:
|Device Name||Model Number|
|Formula 2000 DOMUS||
|Formula 2000 PLUS||
|Formula 2000 PLUS DOMUS||
|Formula Therapy DOMUS||
Dialysis machines are used by patients suffering from chronic kidney failure.
To date, three fires have been reported in three separate machines: one during a cleaning cycle, and two before patient use when the machines were switched ON. In all three cases, no patients or healthcare providers were injured. Investigations have found that the battery charge board was the source of two of the fires, and likely the source of the third.
Bellco is advising all affected customers, via couriered mail, of the risk and providing them with instructions for their service technicians to disconnect the battery charge board to eliminate the possibility of malfunction. Bellco will also be directly contacting providers of homecare patient programs by phone to ensure they have received the necessary instructions. Disconnecting the battery charge board is a temporary solution, and does not affect the performance of the machine. Bellco is working to find a more permanent solution to prevent a possible recurrence.
Bellco will communicate further direction with all customers once a permanent solution is confirmed.
What you should do
- Contact the healthcare provider who administers your dialysis to find out whether you have an affected machine.
- Report complaints about health products to Health Canada by calling toll-free to 1‑800‑267‑9675, or complete an online complaint form.
Stay connected with Health Canada and receive the latest advisories and product recalls using social media tools.
- Date modified: