Important safety information for certain respirator masks
- Starting date:
- May 11, 2020
- Posting date:
- May 11, 2020
- Type of communication:
- Medical Device
- Source of recall:
- Health Canada
- Important Safety Information, Medical Devices
- General Public, Healthcare Professionals, Hospitals
- Identification number:
- Product: Certain respirators, including KN95 respirators.
- Issue: Health Canada has contacted companies that may be importing or distributing certain respirators, including KN95 respirators, in Canada that may not meet safety and effectiveness standards to request that they immediately stop sale and relabel current stock as face masks (not respirators).
- What to do: Provincial and territorial health authorities and healthcare institutions should review their inventories of KN95 respirators to confirm that they meet the Government of Canada technical specifications for healthcare settings for COVID-19 response.
- Products affected
- What you should do
- Media enquiries
- Public enquiries
- What Health Canada is doing
Health Canada is committed to ensuring that the medical devices available to Canadians meet the necessary safety and effectiveness standards. The Department has contacted companies that may be importing or distributing certain respirators, including KN95 respirators, that may not meet expected performance standards in Canada to request that they stop sale and relabel the products as face masks instead of respirators.
This follows the recent U.S. Food and Drug Administration (FDA) communication regarding concerns that certain filtering facepiece respirators from China may not provide consistent and adequate respiratory protection based on testing conducted by the National Institute for Occupational Safety and Health (NIOSH) - National Personal Protective Technology Laboratory (NPPTL) of the Centers for Disease Control and Prevention (CDC).
Health Canada has asked importers and distributors that may have imported respirators that do not meet performance standards to notify their customers and relabel products to indicate that while these masks may not meet the standards required for frontline healthcare workers, they could be used as face masks in settings where a 95% filtration is not needed. The products are not being removed from the market.
This action does not implicate KN95 respirators purchased by the Government of Canada and tested by the Public Health Agency of Canada (PHAC). Before allocating any personal protective equipment to the provinces or territories for frontline healthcare workers, PHAC conducts a quality verification. For KN95 respirators, this includes a visual inspection to check for defects in design and construction, and testing to confirm that they meet filtering specifications. KN95 respirators distributed to provinces and territories by PHAC meet the Government of Canada's technical specifications for healthcare settings for COVID-19 response.
Health Canada is following up with companies that may have imported and distributed respirators that were tested by the CDC NPPTL and did not meet performance standards. Please consult the Medical Device Respirator recalls for a list of products that Health Canada has directed companies to re-label as face masks.
What you should do
- Provincial and territorial health authorities and healthcare institutions should review their inventories of respirators to confirm that they meet the Government of Canada technical specifications for healthcare settings for COVID-19 response.
- Canadians using these masks outside healthcare settings can continue to do so.
- Report any health product adverse events to Health Canada.
What Health Canada is doing
Health Canada will ensure that any companies that have distributed impacted products take appropriate action to stop selling any impacted products, notify customers and relabel existing stock as face masks instead of respirators. Should additional safety concerns be identified, Health Canada will take appropriate action and inform Canadians, as necessary.
- Date modified: